|Gaoxin Industrial Park,Guangming New Zone,Shenzhen City, Guangdong Province, China||Angelwang66@126.com|
|Place of Origin:||China|
|Minimum Order Quantity:||1|
|Packaging Details:||Carton size is determined by the quantity|
|Payment Terms:||T/T, L/C, D/P|
|Amplitude:||0~2000um (peak~peak)||Frequency Response:||4~300 Hz Or 10~500 Hz|
|Sensitivity:||50mV/mm/s ± 5% Or 20mV/mm/s ± 5%||Max. Acceleration:||10g|
|Working Temperature:||﹣30°C ~ +120°C||Dimension:||Φ35×78mm|
DC-CD-6 Magnetoelectric Vibration Transducer in all kinds of vibration measuring
DC-CD-6 Magnetoelectric Vibration Transducer is design and made based on the electromagnetic
induction principle. By this transducer, the vibration signal is converted into corresponding voltage signal.
Because its output voltage proportional to the vibration speed of the machine cover, it is also called
vibration speed transducer.
The double-coil structure is applied in the inside of the transducer, to ensure the superposition of effective
signals, thus improved its interference rejection ability. The sensor has compact dimensions,
excellent sealing performance and long service life, so it is widely used in all kinds of vibration measuring.
Amplitude: 0~2000um (peak~peak)
Frequency Response: 4~300 Hz or 10~500 Hz
Sensitivity: 50mV/mm/s ± 5% or 20mV/mm/s ± 5%
Max. acceleration: 10g
Working temperature: ﹣30°C ~ +120°C
Fixing Screw: M10×1.5mm, Depth: 10mm
Weight: About 200g
Mounting position: transducer is installed horizontally or vertically on the vibration position to be measured,
then fixed with screw M10*1.5 on the bottom.
Installation specification: If the installation position of the transducer is scoured by the high temperature
steam, to reduce the sensor ambient temperature, protective measures need to be taken. No need to take
extra protective measures under normal conditions.
4. Connection between transducer and instrument
Plug and socket on the transducer are marked with number of 1, 2, 3 and 4. In consideration of the secure
connection, “2” is connected with “4” and “1” is connected with “3” already in the inside.
“2, 4” as the output positive end of the transducer signal, it should be connected with input positive end
of the instrument. “1, 3” as the output negative end of the transducer signal, it should be connected with
input negative end of the instrument.
5. Guideline for ordering
DC-CD-6 Magnetoelectric Vibration Transducer is equipped with 1m of cable.
Code of ordering:DC-CD-6 -A□ -B□ -C□
sensitivity A□ : 1 - 20mV/mm/s
2 - 50mV/mm/s (*)
frequency response B□ : 1 - 10~500 Hz (*)
2 - 4~300 Hz
Direction of measuring C□ : 1 - V & H(*)
2 – V (4~300 Hz)
3 – H (4~300 Hz)
DC-CD-6 -A2 -B1 -C1
Assorted Monitoring Instruments :
Instruments matched with DC-CD-6 Magnetoelectric Vibration Transducer is as below:
-- DM-ZD-W/L Disk-type Vibration Monitoring Instrument
-- DM-ZD-WG/LG Wall-type Vibration Monitoring Instrument
Weihai grand exploration instrument Co., Ltd
Add: No.455 Shichang Road, Weihai City, Shandong Province, China